Edwards APG100-XLC Corrosive Resistant Active Pirani Vacuum Gauge,KF25 Flange
The Edwards APG100-XLC Corrosive Resistant Active Pirani Vacuum Gauge contains a NW25, KF25 vacuum flange, is compact for easy installation, has a linear output, and a user replaceable tube sensor. The new Edwards gauges are compatible with all Edwards TIC, ADC, TAG instrument controllers and other active gauge controllers and displays. They are also CSA, C/US approved as well as fully RoHS compliant due to their lead-free construction. They measure pressure from 1000 down to 10-4 mBar. The signal cables and TIC vacuum gauge controllers are sold separately. The instruction manual for these Edwards APG100 series gauges is available in PDF format below. The APG100 is available in two versions: the ‘M’ version (contains standard Tungsten/Rhenium filaments) can measure pressure down to 10-3 mBar and is suitable for general applications; the ‘LC’ version (contains corrosive resistance Platinum/Iridium filaments) can measure pressure down to 10-4 mBar and is also suitable for use in corrosive applications.
The APG100 can be mounted in any orientation however the gauge tubes are individually factory calibrated in nitrogen whilst vertical. For correct pressure indication in your chosen gauge orientation, the gauge should be recalibrated at atmospheric pressure. Edwards recommends mounting the gauge tube vertical in order to minimize the build up of process particulates and condensable vapors within the gauge. he APG100 is calibrated for use in nitrogen, and will read correctly with dry air, oxygen and carbon monoxide. For any other gas type a conversion is required in order to obtain the correct pressure reading, common gases: nitrogen, argon, carbon dioxide, helium, krypton and neon.
- Easy Access to signal cable with compact enclosure
- Sensor bakeable to 150°C (300°F) and sensor is user replaceable
- Adjustable set point for simple process control and interlocking
- Remote calibration possible
- Pressure range 10-4 mBar
- Linear output – 1 Volt per decade for easy interface with vacuum control systems
|Internal volume||5 cm3|
|Measurement range (APG100-XM)||Atmosphere to 10-3 mbar|
|Measurement range (APG100-XLC)||Atmosphere to 10-4 mbar|
|Accuracy (APG100-XM)||Typically +/- 15% at|
|Accuracy (APG100-XLC)||Typically +/- 15% at|
|Maximum over-pressure||10 bar absolute|
|Operating temp range||5o to 60o C|
|Storage temp range||-30o to 70 oC|
|Bake-out with no electronics||150 oC|
|Humidity||80% RH up to 31 oC decreasing linearly|
|to 50% RH at 40 oC and above|
|Maximum altitude||3000 m|
|Filament temperature||100 oC above ambient|
|Electrical supply voltage||15 to 30 V d.c. nominal|
|13.5 V d.c. minimum|
|32 V d.c maximum|
|Power consumption||1 W|
|Output signal||0 to 10 V d.c. nominal|
|Set-point – open collector transistor|
|Rating||30 V d.c. 100 mA|
|Range of set-point||1.8 to 9.2 V dc|
|Fixed hysteresis||500 mV (1/2 decade)|
|Level setting resolution||6 mV|
Pirani Gauge for Measuring Vacuum
The Pirani Gauge is used to measure low vacuum. In the system tungsten filaments are heated up by passing current through them. As gas molecules interact ...
Cleaning of Edward Pennig gauge
In this Video I am discussing about the cleaning of Edward's Penning gauge. Normally after using sometime these gauges do not work because discharge does ...
The videos are from Youtube, if there is any infringement, we will delete them, please send an email to firstname.lastname@example.org, thanks.