Edwards APG100-XM Active Pirani Vacuum Gauge, KF16 Flange
The Edwards APG100-XM Active Pirani Vacuum Gauge contains an NW16, KF16 vacuum flange, is compact for easy installation, has a linear output, and a user replaceable tube sensor. The new Edwards gauges are compatible with all Edwards TIC, ADC, TAG instrument controllers, and other active gauge controllers and displays. They are also CSA, C/US approved as well as fully RoHS compliant due to their lead-free construction. They measure pressure from 1000 down to 10-3 mBar. The signal cables and TIC vacuum gauge controllers are sold separately. The instruction manual for these Edwards APG100 series gauges is available in PDF format below. The APG100 is available in two versions: the ‘M’ version (contains standard Tungsten/Rhenium filaments) can measure pressure down to 10-3 mBar and is suitable for general applications; the ‘LC’ version (contains corrosive resistance Platinum/Iridium filaments) can measure pressure down to 10-4 mBar and is also suitable for use in corrosive applications. These Edwards APG100-XM Active Pirani vacuum gauges with standard Tungsten/Rhenium filaments and NW16 vacuum flange are new with Edwards part number D02601000.
The APG100 can be mounted in any orientation however the gauge tubes are individually factory calibrated in nitrogen whilst vertical. For correct pressure indication in your chosen gauge orientation, the gauge should be re-calibrated at atmospheric pressure. Edwards recommends mounting the gauge tube vertical in order to minimize the build-up of process particulates and condensate vapors within the gauge. he APG100 is calibrated for use in nitrogen, and will read correctly with dry air, oxygen and carbon monoxide. For any other gas type conversion is required in order to obtain the correct pressure reading, common gases: nitrogen, argon, carbon dioxide, helium, krypton, and neon.
Edwards Presents Vacuum Gauges
An Introduction to the Edwards range of Active, Passive and Digital Gauges and Controllers.
Cleaning a high-vacuum Penning gauge (cold cathode vacuum gauge)
My high-vacuum Penning gauge has been having some problems lately. The last time that I "fixed" it, I used lead solder to create a crushable metal seal ...
The videos are from Youtube, if there is any infringement, we will delete them, please send an email to [email protected], thanks.